The Boat Anchor Manual Archive Path: /hp/a/. This is the BAMA archive. These manuals are available for download and free of charge. I am now accepting. View and Download HP A operating and service manual online. DIGITAL VOLTMETER. A Measuring Instruments pdf manual download. Hp A Pdf User Manuals. View online or download Hp A Operating And Service Manual.
|Published (Last):||14 March 2007|
|PDF File Size:||20.66 Mb|
|ePub File Size:||17.69 Mb|
|Price:||Free* [*Free Regsitration Required]|
At each setting, the A reading should be within manuql Test Limits given in the table. Contact the seller – opens in a new window or tab and request a shipping method to your location.
Front-Terminal Reeding Step I: Help me help you! If the problem cannot be cor- rected by the adjustment, refer to Section VIII for troubleshooting information. Disconnect the DC Transfer Standard.
The measurement result is stored by the main controller as the 10 V dc full scale constant. Auto-Cal Constant 9 10 K. Derive 6-month test limits from AC Accuracy specifications listed ph Table Table gives the typical measurement times, using the HP-IB.
Full text of “HP: a”
NMR is the ratio of the peak normal-mode voltage to the peak error voltage in the reading. The references are switched at a 1 milli- second rate during the integration period milli- seconds. With the A set in the Scale mode, the value of the unknown resistor is displayed without the input lead resistance. The Model A measures dc voltage from 1 dummy cal constant calculation made in the 33455a microvolt to KXX volts in five ranges extending from.
Each key has an LED which lights to indicate the trigger source 3545a.
Record the 34S5A reading:. The measurement time on each range should be added to the total measurement time. DC and Ohms Reference Adjustments. Intemipt end Device Status Massages.
The output of the operational attenuator main controller as the lOO V dc gain error constant. This transfers 3455z number from the storage register specified to the display. The full- scale accuracy is therefore: HI Acbisrut 2K For rated measurement accuracy, the SA should be allowed to warm up for at least one hour.
During the warranty period, Hewlett-Packard Company will, at its option, either repair or replace products which prove to be defective. Interrupt and Rptriro Require Service Indicates a device’s need for inter- action with the controller.
If the original container is avail- able. Please enter a valid ZIP Code.
Add to watch list. The trigger input is to be driven with TTL level signals.
Set the Decade Resistor to K ohms. The range selected by the instrument is the I V range. If this is done, check the full-scale accuracy of the 34S5A 1 V and 10 V ranges using the Reference Divider recommended in the procedue.
The test equipment required for the performance tests is listed at the beginning of each procedure and in the Recommended Test Equipment Table in Section I. This voltage drop is proportional to the current through the resistor. Equipment Required Also listed are available options and accessories, and instrument and manual identification information.
To read control status: DC Operational Accuracy Test ECMR is the ratio of the peak common-mode voltage to the resultant peak error voltage in the reading with I k l un- balance in low lead. Various examples using the scale mode are as follows: Instrumont Mtaauroffloiit Timoi Remots Control.
Basic Ohms Measurement Diagram. Jumper Removal A15 board.